Difference between revisions of "Fluorescence Optical Detector Board"
|Line 30:||Line 30:|
== Luminex ==
== Luminex ==
Revision as of 21:59, 10 September 2020
Learning from reverse engineering the Illumina Optical Detector Board extracted from a NeoPrep system (discontinued).
The device electronics is quite straight forward. There is one line of 15 blue LEDs for excitation and two lines of photodiodes. The LEDs are controlled by an LED driver ICs and the photodiodes are read by an photo diode ADC. All comes together by a FPGA (black box). The power pins and input lines to the module are easy accessible on colour coded probe pins (nice).
The 3 colour filters are:
Filter blue (excitation): 450 - 480 nm
Filter green (emission) : 510 - 550 nm
Filter red (emission) : 600 + nm
Testing the sensor with "R&D Systems ® Luminex ® Bead-Based Assays for Multiplexing" kits.
There are two light sources used to collect data from samples. The red light source (laser for Luminex® 100/200™ and LED for MAGPIX®) excites the internal dyes in the microspheres so that they can be classified on the bead map. The green light source excites any reporter bound to the surface of the beads (PE-Streptavidin).
In Luminex applications, SA-PE is excited by the 532 nm green laser or 520 nm green LED.
Its emission peak exists at 575 ± 10 nm (orange/yellow).
Systems with two LEDs:
(511 nm) green
(621 nm) red
Excitation: 488 nm blue (azure) (broad excitation, one peak 488 nm, one at 575 nm)
Emission: 575 nm (yellow)
Excitation 621 nm red ?
Emission? two dyes for fluorescence? Infrared?